To compare two angiography systems of different image capture technology, one with flat detector (FD) and one with image intensifier (II), in terms of entrance surface air kerma (ESAK) rate, detector dose (DD) rate and image quality (IQ), in interventional cardiology procedures concerning both adult and pediatric patients.
Materials and methods
In order to determine ESAK and DD rates, a digital dosimeter and polymethylmethacrylate (PMMA) plates were used. For the evaluation of IQ, two contrast objects (the Leeds TOR 18FG and a 5 mm-thick Aluminum plate) were used and two figures of merit were defined in fluoroscopy and cine acquisition modes. Measurements of ESAK, DD rates and IQ were made for various fields of view, pulse and frame acquisition rates.
For the particular setup used in this study was noted that ESAK values in the II system were generally larger than the respective values in the FD system (on average 70% for fluoro mode, 5 times for cine mode). When halving the fluoroscopy pulse rate, reduction in ESAK was not proportional, in fluoroscopy mode. Image quality evaluations indicated that II performs better in terms of low contrast sensitivity (LCS) and signal-to-noise ratio (SNR) than the FD system which performs better regarding high contrast resolution (HCR). However, when considering image quality in relation to ESAK the FD system performs better than the II system (with the exception of low thicknesses and zooms for high pulse rates in the fluoroscopy mode).
The FD system, generally, provides a better image quality–dose relation than the II system although II unit provides better LCS and SNR. This means that with the right adjustments to both systems, FD unit is able to provide same image quality with lower dose. However, newer technology does not automatically imply better image quality and further investigation is necessary for deriving safe conclusions for units which utilize different capture technology.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Physica Medica: European Journal of Medical Physics
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Dual-energy cardiac imaging: an image quality and dose comparison for a flat-panel detector and x-ray image intensifier.Phys Med Biol. 2007; 52: 183-196
- Strategies to improve the signal and noise performance of active matrix, flat-panel imagers for diagnostic x-ray applications.Med Phys. 2000; 27: 289-306
- Antonuk L.E. Yaffe M.J. Design and performance of a high quality cardiac flat detector. Proceedings of medical imaging 2002: physics of medical imaging; 2002 February 24; San Diego, USA. The International Society for Optical Engineering, Washington, USA2003
- Antonuk L.E. Yaffe M.J. Image quality of a large-area dynamic flat detector: comparison with a state-of-the-art II/TV system. Proceedings of medical imaging 2002: physics of medical imaging; 2002 February 24; San Diego, USA. The International Society for Optical Engineering, Washington, USA2003
- Radiation dose and image quality for pediatric interventional cardiology.Phys Med Biol. 2008; 53: 4049-4062
- Does digital flat detector technology tip the scale towards better image quality or reduced patient dose in interventional cardiology?.Eur J Radiol. 2009; 72: 348-353
- Comparative evaluation of an II based and a flat panel based cardiovascular fluoroscopy system within a clinical environment.Australas Phys Eng Sci Med. 2005; 28: 151-158
- Dynamic flat panel detector versus image intensifier in cardiac imaging: dose and image quality.Phys Med Biol. 2005; 50: 5731-5742
- Do flat detector cardiac x-ray systems convey advantages over image-intensifier-based systems? Study comparing x-ray dose and image quality.Eur Radiol. 2007; 17: 1787-1794
- Flat-panel detectors: how much better are they?.Pediatr Radiol. 2006; 36: 173-181
- Radiation exposure during elective coronary angioplasty.Heart Lung Circ. 2008; 17: 215-219
- Dose performance evaluation of a charge coupled device and a flat-panel digital fluoroscopy system recently installed in an interventional cardiac laboratory.Radiat Prot Dosim. 2004; 111: 297-304
- Comparison of a CCD and a flat-panel digital system in an interventional cardiology laboratory.Radiat Prot Dosim. 2005; 117: 93-96
- Part 2–43: particular requirements for the safety of x-ray equipment for interventional procedures.([IEC 60601-2-43])2nd ed. 2010
ICRU. report 74: Patient Dosimetry for X Rays Used in Medical Imaging. 2005; 5(2).
- The optimization of signal detectability in digital fluoroscopy.Med Phys. 1982; 9: 688-694
- Signal detectability in digital radiography: spatial domain figures of merit.Med Phys. 2003; 30: 2180-2193
- Assessment of performance of a new digital image intensifier fluoroscopy system.Radiat Prot Dosim. 2008; 129: 123-126
- Are new technologies always reducing patient doses in cardiac procedures?.Radiat Prot Dosim. 2005; 117: 97-101
- Paediatric interventional cardiology: flat detector versus image intensifier using a test object.Phys Med Biol. 2010; 55: 7287-7297
Published online: March 07, 2012
Accepted: February 1, 2012
Received in revised form: December 10, 2011
Received: May 19, 2011
© 2012 Associazione Italiana di Fisica Medica. Published by Elsevier Inc. All rights reserved.