- •Monte Carlo implementation of a micro-CT scanner (XtremeCT)
- •Validation of MC model using measurements.
- •Extensions of MC model to enable the analysis of scatter components.
- •Scatter component analysis for different situations.
Micro computed tomography (µCT) scanners are used to create high-resolution images and to quantify properties of the scanned objects. While modern µCT scanners benefit from the cone beam geometry, they are compromised by scatter radiation. This work aims to develop a Monte Carlo (MC) model of a µCT scanner in order to characterize the scatter radiation in the detector plane.
The EGS++ framework with the MC code EGSnrc was used to simulate the particle transport through the main components of the XtremeCT (SCANCO Medical AG, Switzerland). The developed MC model was based on specific information of the manufacturer and was validated against measurements. The primary and the scatter radiation were analyzed and by implementing a dedicated tracing method, the scatter radiation was subdivided into different scatter components.
The comparisons of measured and simulated transmission values for different absorber and filter combinations result in a mean difference of 0.2% ± 1.4%, with a maximal local difference of 3.4%. The reconstructed image of the phantom based on measurements agrees well with the image reconstructed using the MC model. The local contribution of scattered radiation is up to 10% of the total radiation in the detector plane and most of the scattered particles result from interactions in the scanned object. The MC simulations show that scatter radiation contains information about the structure of the object.
In conclusion, a MC model for a µCT scanner was successfully validated and applied to analyze the characteristics of the scatter radiation for a µCT scanner.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Physica Medica: European Journal of Medical Physics
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Micro-CT of rodents: state-of-the-art and future perspectives.Phys Med. 2014; 30: 619-634
- The Zurich experience: one decade of three-dimensional high-resolution computed tomography.Top Magn Reson Imaging. 2002; 13: 307-322
- Assessment of residual error for online cone-beam CT-guided treatment of prostate cancer patients Int.J Radiat Oncol Biol Phys. 2005; 62: 1239-1246
- Cone-beam-CT guided radiation therapy: a model for on-line application Radiother.Oncology. 2005; 75: E1-8
- In vivo determination of bone structure in postmenopausal women: a comparison of HR-pQCT and high-field MR imaging.J Bone Miner Res. 2008; 23: 463-474
- Accuracy of high-resolution peripheral quantitative computed tomography for measurement of bone quality.Med Eng Phys. 2007; 29: 1096-1105
- Characterization of scattered radiation in kV CBCT images using Monte Carlo simulations.Med Phys. 2006; 33: 4320-4329
M. Endo, T. Tsunoo, N. Nakamori, K. Yoshida, Effect of scattered radiation on image noise in cone beam CT, Med. Phys. 28, 469-474.
- Comparison of EGS4 and MCNP4b Monte Carlo codes for generation of photon phase space distributions for a Varian 2100C Phys.Med Biol. 1999; 44: 3009-3026
- A multiple source model for 6 MV photon beam dose calculations using Monte Carlo.Phys Med Biol. 2001; 46: 1407-1427
- Monte Carlo calculation of nine megavoltage photon beam spectra using the BEAM code Med.Physics. 2002; 29: 391-402
- Sensitivity of megavoltage photon beam Monte Carlo simulations to electron beam and other parameters Med.Physics. 2002; 29: 379-390
- Clinical implementation of a Monte Carlo treatment planning system Med.Physics. 1999; 26: 2133-2143
- An efficient framework for photon Monte Carlo treatment planning.Phys Med Biol. 2007; 52: N425-N437
- Clinical use of a commercial Monte Carlo treatment planning system for electron beams.Phys Med Biol. 2005; 50: 1029-1034
- A flexible environment for Monte Carlo simulations: study of inhomogeneities and phantom shape in brachtherapy.J Phys: Conf Series. 2007; 74: 021022
- Implementation of amorphous silicon detector geometries in a flexible Monte Carlo environment.J Phys: Conf Series. 2007; 74: 021005
- Dose response of an a-Si: H EPID on static and dynamic photon beams.Med Phys. 2002; 29: 1269
- Assessing dose rate distributions in VMAT plans.Phys Med Biol. 2016; 60: 3208-3221
- Experimental validation of rapid Monte Carlo based micro-CT simulator.Phys. Med. Biol. 2004; 49: 4321-4333
- The EGSnrc code system: Monte Carlo simulation of electron and photon transport, Technical Report PIRS-701.National Research Council of Canada, Ottawa, Canada2000
I. Kawrakow, egspp: the EGSnrc C++ library, Technical Report PIRS-899, National Research Council of Canada, Ottawa, Canada, 205.
- BEAMnrc Users Manual, NRCC Report PIRS-0509(A)revK.Ottawa, Canada2009
- The EGSnrc C++ class library, NRC Report PIRS-898 (rev A).Ottawa, Canada2009
- Flat panel detectors for digital radiography.in: Handbook of Medical Imaging. Vol. 1. SPIE Press, Bellingham, WA2000: 223-328
Accepted: November 4, 2017
Received in revised form: October 5, 2017
Received: June 10, 2017
© 2017 Associazione Italiana di Fisica Medica. Published by Elsevier Ltd. All rights reserved.